This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
I want to asking about Op Amp specification to realize thermal noise measurement on conductor. What the most important Op Amp specification for that application?What the specification for measure noise voltage and noise current is different?Can you give me example the Op Amps type which are appropriate for that application?
Whenever you use an op amp to measure the noise of something, the most important paramaters are:
The OPA211 is an example of a device with very low voltage spectral density (1.1.nV/rtHz). The current noise for this device is not optimized (3.2pA/rtHz).
The OPA827 has good voltage noise 4nV/rtHz, and good current noise 2.2fA/rtHz.
I highly suggest going through the noise precision labs videos. There are 9 videos and they cover a lot of detail on this subject.
Art Kay Senior Applications Engineer High Performance Linear
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Art Kay:
In reply to Saaddin Nur:
A common method used to measure low noise circuits is discribed in the TI design given in the link below. The key is to follow your amplifier with low noise amplifier that will boost the noise signal to a level you can measure with a scope. If you provide a schmatic of your circuit we can simulate this. You should definately sim it before you build the measurement circuit.
Attached is a good presentation that covers the subjects of chassis grounding, and PCB grounding. I think slides 39-41 are most pertinent to your question. I suggest looking through the slides and see if they help. Henry Ott has a very good book on the subject also “/Electromagnetic Compatibility Engineering”.
I read some of your OP Amp noise presentation. and learn that we use some gain and post gain amp to get the Amps noise plot in 0.1-10hz.
but, I'm confuse that, such like in OPA333 datasheet page1, there is a noise plot, the resolution is about 500nV/div, how we can get such low resolution of oscilloscope?
In reply to Haroad:
Regards,Collin WellsPrecision ADC Applications
In reply to Collin Wells:
Just to add to what Collin said. We have a TI Precision design reference design that covers this. See link below
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.