# TX 517 EVM Power Supplies - # of external supplies, max. slew rate

In a previous post it was confirmed that the voltages HV1, LV1, HV0, LV0, HV2, LV2 + VDD, VEE, VCM must be supplied externally to the evaluation kit.

- the max power supply slew rate should be 10V/ms (page 9)
- the external supplies need to be sequenced according to table 4, page 14
power up
1) Driver supplies
2) LV1,
3) HV1,
4) LV2, LV0, HV0, HV2, VCW

and the reverse for power down

According to the user manual for the EVM, the user of the eval kit has to externally supply the following 10 voltages:

VDD = 5V
VEE - 5V
HV0 = +1.9V
LV0 = -1.9V
VCW = 11V
VAA = 2.5V
HV1 = 61V
LV1 = -20.9V
HV2 = 32V
LV2 = -11.9V

The EVM only generates
- 3.3V using the TPS79633DCQR
- 1.2V using the TPS73201QDBV

So my questions are
1) The user has to apply 11 external voltages to get the EVM kit functional?
a) 5V supply for the EvalKit
b) 10 Voltages VDD, VEE, VCW, VAA and HVx, LVx

2) What are the consequences (for the TX517) if the power supplies used exceed the max. power supply slew rate requirement of 10V/ms as specified for the TX517 ?

thank you
Oliver

### 4 Replies

• Oliver,

Thank you for your interest in the TX517. I am assigning your post to the correct applications engineer. He should respond soon.

Amy Schnoor

HealthTech

Mixed Signal Applications

Our TI Medical E2E Forum is a large and dynamic support community!

• Oliver,

In general, yes, they are all required for full functionality of the device.  If CW mode is not used, then VCW is not required.  HV0 and LV0 are used for damping the output at low voltages, so it only helps the performance.  To get 17 levels, then two channels of 5-level output are required.  You can simply use the HV1/LV1 and HV2/LV2 as required for your application and also scale the voltages to your application. The driver voltages are always required.

Thanks,

Chuck Smyth

Chuck Smyth, Applications Engineering, HealthTech-Imaging