This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
Part Number: TPA3136D2EVM
I am working on the audio demo using the TPA 3136D2 EVM. The load is a
PBTL mono load.
I am using a 4 Ohm 3W resistor for this, resistance measures 4 Ohm on a multimeter.
This resistor is connected across the OUTL+ and OUTR+ points on the EVM.
I have a 1 kHz, 480 mV p-p signal at the RINP TP of the EVM. I can see this on the scope.
I have the voltage between PLIMIT and GND set to 1.25 V.
I am powering the EVM at 12V DC.
I cannot see anything on the scope when I connect a probe across the resistor ends.
I am unable to read any voltage across the resistor leads as well with a multimeter,
AC or DC.
The DC voltage between OUTL+ and GND measures 5.9V DC. Same voltage between
OUTR+ and GND.
Can you offer any pointers on this? I have attached the EVM configuration used
for a mono PBTL load. Does this need changing?
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Adam Sidelsky:
In reply to Priya Nadathur70:
The first picture is the input to the EVM, 1kHz, 490 mV signal. Two waves are seen because one is RINP,
the other is the function generator output.
The loudspeaker when connected gives a loud tone, no clicking/ticking heard like from the past.
The scope picture however is noisy, only the left part of the speaker shows a switching waveform,
right part is 0 V (maybe this is how the mono output looks like?)
I was hoping to snap a perfect EVM output waveform picture.
In reply to Shawn Zheng:
Attached are 3 pictures of the test bench setup:
1. Wavetek function generator provides 1 kHz sine wave input, 490 mV p-p.
2. EVM shown with input connected to RINP. Note that the PBTL jumper is the only jumper in
the EVM that is open.
3. An RCA cable adapter is attached to the function generator probe. One end of this RCA cable is
used for the input connections, the sleeve of the other end of this cable is used for the GND connections.
Please review and offer your feedback ASAP. If the configuration needs changing, we need to know right away.
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with respect to these materials. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.