This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
Part Number: ADS7223
we are using the ADS7223 in full clock mode. Our idea was to read the CID with each conversion for plausibility testing. It seem that in this mode the channel number reported is always "00" when CID = "0" which normally should report the Channel ID as per the description of BIT5 (CID disable) on page 35 of the datasheet. There is a hint above the diagrams on page 27 of the datasheet.
However, is there any way to read the CID when using full clock mode? Or is this feature simply not available in full clock mode?
Hi Lutz, Thank you for your post. Can you check whether the customer is using pseudo- or fully-differential inputs (PDE = 0 or 1)? Also, how is the customer selecting the channel - manually or automatically (M0 = 0 or 1)? In Table 4, it looks like Channel Information is only not available when PDE = 1 and M0 = 1.
Applications Engineer | Precision ADCs
Are you working on a bio-potential application? Check out these helpful resources:
Common Questions for TI's ADS129x Family of Bio-Potential ADCs
How to Measure ECG: A Guide to the Signals, System Blocks, and Solutions
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Ryan Andrews:
Hi Ryan, it is pseudo diffetential input mode andnthebchannelsnswlxted manually. We would like to know if switchong the channels has happened.
In reply to Lutz Naumann:
Ok, so PDE = 1 and M0 = 1? I'll have to talk with Tom Hendrick about what it could be. I'll get back to you soon. Regards,
It is Mode 1 and addressing the next channel manually, Pseudo-Diff on all channels and both SDOx are separate to get the speed up to 1MSPS.
this is what we were expecting as well. Unfortunately in our case the MSBs don't report what channel is being sampled. According to the designer we get 00. We are not sure if we are doing something wrong.
Sorry for the delay. We have an ADS8363EVM (same interface) that we are going to bring up in the lab today.
Meanwhile, can you share a capture of the interface and tell us the clock frequency and data rate?
the part is operated in full clock mode, Mode I, with 1MSPS per ADC. 12bit plus the 2 CID Bits. I do not have the exact SPI clock frequency. But for the functionality of the CID it should not matter from my understanding As this is more about the content of the frame.
Posting a summary of our offline conversation here to close this out:
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.