ADS1282: calibrate gain ads1282
Part Number: ADS1282
I'm planning on using ADS1282 on an Eval board, in bipolar mode for Gain Cal, I plan to use +/-2.5V on VREFP/N & +/- 2.5V on AVDD/AVSS. (Vref = VREFP – VREFN) which should be Vref = 2.5 - (-2.5) =5V with PGA =1, is that correct ?
In Bipolar mode with a 5V Vref (as above) for gain cal must supply input signal pins should be +1.25V to INP and - 1.25V to INN that correct? +/-Vref/(2*1) = 2.5/2 = 1.25V
in Unipolar I plan to have 5V on VREFP & 0V on VREFN, is that ok?
Welcome to the TI E2E Forums!
Correct...The ADS1282 will take a 5V (differential) reference voltage and measure input signals up to +/- 2.5V (differential). Therefore, with a 5V reference, your full-scale input signal for gain calibration will be 2.5V.
VREFN is typically biased to AVSS, so in bipolar mode AVSS = VREFN = -2.5V; while in unipolar mode AVSS = VREFN = 0V.
Best regards,Chris HallApplications Engineer | Precision ADCs
Check out these helpful resources...TI Precision Data Converters | TI Precision Labs - ADCs | Analog Engineer's Calculator | Data Converters Learning Center | Selection Guide
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Christopher Hall:
Thanks for confirming my differential set-up is correct.
For the Unipolar config with AVDD = VREFP = 5V, AVSS = VREFN = 0V, would AIP = 2.5V & AIN = 0V, is that correct?
In reply to Chris Sanderson:
There is an "absolute input range" requirement on the ADS1282 that says that AINP and AINN should not exceed (AVDD - 1.25V) or drop below (AVSS + 0.7V). ...This is another way of specifying an input common-mode range requirement.
The differential 2.5V is okay, but the common-mode voltage would out of specification if AINN < 0.7V.
I would recommend setting AINP = 3.475V, and AINN = 0.975V to bias the 2.5V differential signal to the center of the PGA's input common-mode range; however, any 2.5V differential voltage that meets the "absolute input range" requirement will work.
So for Unipolar I plan to use +5V on AVDD/VREFP & 0V on AVSS/VREFN and for the input pins, AINP = 3.475V, and AINN = 0.975V.
Please confirm that is correct.
That is correct.
One last question hopefully.
For the Unipolar input pins, AINP = 3.475V, and AINN = 0.975V can we change this to AINN = 3.475V, and AINP = 0.975V? is that a concern.
It's not a problem to swap AINP and AINN, but you'll have to manually perform the calibration as outlined in section 9.30 of the datasheet. When computing equation 11, the negative full-scale code will be your expected output code.
If you send the GANCAL command the device is going to expect a positive full-scale signal. Running GANCAL with a -2.5V differential input may produce some odd results in the FSC[2:0] registers.
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.