This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
I want to design a Doppler circuit with two AFE5809s, and use 16 piezo elements to realize a Doppler measurement. The analogue to numerical conversion, is realize with an ADS8472 like the AFE5809 datasheet proposed (‘’CW Supporting Circuits’’ section). In attachment, you can find my proposed circuit (multiple AFE in application note).With this circuit; I have a couple of question.
1 : Can you say me if my proposed circuit is good? And if is not can you give me the good one?
2 : If my circuit is ok, What is the routing constraints of the differentials pairs? Does the diff pairs of CW_IPAFE1_OUTM, CW_IPAFE1_OUTP, CW_IPAFE2_OUTM, CW_IPAFE2_OUTP, CW_QPAFE1_OUTM, CW_QPAFE1_OUTP, CW_QPAFE2_OUTM and CW_QPAFE2_OUTM must have the same lengths?
Finally, we have pay the AFE5809 dev board, can you say me if can i have the Gerber file or eventually the Altium, Cadence files...of this AFE5809 dev board? Can you send me one of them?
Your configuration is correct for the CW Voltage Output mode (Figure 103). Also, see the Design database for the AFE5809 EVM here:
Do you still need the Gerbers?
Chuck Smyth, Applications Engineering, HealthTech-Imaging
Please verify post if question is answered.
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to csmyth:
Hi ChuckThanks a lot.Yes i want the GERBER file (i can read it without difficulty).For the Design database (AFE5809 EVM), can you say me what software can i use to read it.Best regardsSam
In reply to Samm:
Thanks for the Gerbers file
Hi Chuck,In attachment, you can find my final design. Before routing it, i want to confirm with you my design.
For the values of R5,11,16,18,25,31,36 and finally R38, i chose these values equal to 499 Ohms. Does these values are good?
Finally, on the ti website, i find the AFE5816. Can you say me more about it the price, datasheet, and eventually dev board.
Please see the attachment for the CW Mixer Out external amplifier reference. I recommend an external amp, but I have not tried going straight into the ADC. In this case, I believe the 500 ohm might be unnecessary.
CW Mixer Out from AFE5809_EVM_RevA_Sch.pdf
Can you please provide me detailed datasheet for AFE5816 under NDA condition. I can email details about my application
Vassili Ivanov, PhDSystem Architect / CTO / co-founderPhotoSound Technologies, Inc
In reply to Vassili Ivanov40:
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with respect to these materials. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.