I have used the ADS1230 in my loadcell application.in which i have used the CY8C29666 -24PXI MCU,16X2 LINE LCD & ADS1230 (ADC),.
I am facing the display problem,that what ever the weight indicated on display get zero & ADC get heng up.when i start/stop the 230VAC contactor in electrical panel.
i have used 5 v dc LDO for the loadcell supply & MCU.i have used RC filter 0.1u/270vac Cap + 100ohms resistor across the contactor ,but there in no improvement in display.
so please give me some suggestion to avoid this problem.
i am attaching the sch.of ADS1230 ,
If the ADS1230 device gets 'hung up' when you turn on/off your contactor, there is probably a surge either in the 5V rail, or a differential in the ground. How do you have your board layout configured? Do you have a solid ground plane or are you point-to-point wired? I don't see the 5V LDO in your schematic, what sort of filtering do you have there and how much bulk capacitance do you have on the 5V rail?
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with respect to these materials. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.