Part Number: DLPNIRNANOEVM
We just use CCS to compile the Mobile Spectroscopy TIVA EVM onto NIRNANO scanner with XDS100v2 probe connected. But during the process, the PC cannot recognize the NIRnano. When we try to use reset button to reset the NIRnano, the power LED is on all the time, which is not the right way it should be (blink). Could you please let me know how to reset the NIRnano with either JTAG or NanoSD card? Thanks for your help.
In reply to Philippe Dollo:
We figure out how to reset the NIRscan Nano. But we found new problems with now, after we load the "NIRscanNano.out" to the EVM, the intensities reading is very abnormal (I attached here).
I need to update Tiva Firmware "NIRscanNano.bin" to make the reading normal.
We have to upload the original one download from TI, not the one generated on my PC, when I built the project. Could you give me some suggestion on this? Appreciate your help.
In reply to YANJUN LI:
In reply to Danny White:
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with respect to these materials. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.