This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
Part Number: BQ34110EVM-796
now I use the BQ34110EVM-796 to calibrate,however during the calibration,the bqstudio report error,and the error detail information is "Calibration - pack voltage correction is beyond allowed range;
I use the NI-MH batterry and the voltage is 10V and the calibration flow is referred to the TRM,now I don't know how to resolve it.
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Matt Sunna:
In reply to darren:
Did you follow all of the steps in the EVM User Guide for the setup? What are you setting the Voltage divider to on the Calibration screen? It should be about 19200 based on the divider you are using.
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.