Part Number: ADS58J89
I have a ADS58J89 hooked up to a TSW14J65.
I followed the instructions provided in the datasheet of the both modules to install HSDC pro (version U) and ADS58J8x EVM GUI (Version A).
First I clicked on device reset on ADS58J88 Tab on the ADSJ58J8x GUI. Also, I clicked on reset for LMK04828 PLL1 and PLL2.
There are several problems.
1- On HSDC pro, when I click on select ADC, I cannot find ADS58J8x_LMF_442. I only can find ADS58Jx_LMF442. Is that ok?
2- On ADS58J8x EVM GUI, ADS58J88/JESD configuration I put F=1, K=32 and L=4
In the user manual of ADS58J8x EVM, on the section titled "JESD204B controls" says the capture card must be configured with same parameters of ADS58J89. How to put same values in the capture card?
3- On the low level tab in ADS58J8x EVM GUI, load config, there are two options : ADS58J88_500M_442.cfg , and ADS58J88_500M_841.cfg , but the user manual says there must be ADS58J88_500M_442.cfg and ADS58J88_500M_881.cfg . What is this difference for? I want to sample with Fs=250M.
4- After loading either ADS58J88_500M_442.cfg , or ADS58J88_500M_841.cfg, here is status of my LEDs on ADS58J88:
D1 : on
D4 : on
I tried to see SYSREF signal by connecting either J4 or J17 to an Oscope/spectrum analyzer, but I couldnt see any signal. (This state is before checking the SDCLKout_PD in block of controls for clocks CLKout2 and 3, which turns SYSREF off)
5- I put Fs=250M on HSDC pro. and pushed on Capture, but I got an error. I have attached a screen shot of it here.
I followed the instructions and here is status of the LEDs: D4 on TSW14J56 is not blinking. D3 on ADS58J8x is on!
Would you help me to make it work?
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Ebenezer Dwobeng:
Thanks for the reply.
I used the new firmware in HSDC pro, but a new error popped up. Here it is:
Also, still D4 in capture card is off and not blinking.
The more important point is I monitored J4 and J17 (SDCLKOUT7) by an oscilloscope, but it's not showing anything. I think the ADC is not sending any sync signal out.
How can I solve this issue?
In reply to Farshid Tamjid:
Can you confirm the current limit of power supply to TSW14J56 is more than 3A? I am also putting together a setup in the lab with the same version of software you have to debug further if increasing the current limit does not help.
Hi Eben, Well both the ADC and capture card came with their own power supply. They both have PHIHONG swtiching power supply Model: PSC15A-050 OUTPUT: 5V, 3A Regards Farshid
The right .ini file for ADS58J89 was not included in v5p0 of HSDCPRO. I will have them added to the next release.
But you can use the ini file ADS54j54_LMF_442 also for your testing because this device JESD204B interface is identical to ADS58J89.
Thanks for the help. It started working when I chose ADS54J54.
I have another question too. How can we export real-time data from four channels of ADS58J89 and import in Matlab? Does TI have any interface for it?
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.