Part Number: LDC1101
The average test value of the inductance with GUI is 6.19836uH on EVM user's guide Page17, so I'd like to know if this indicates the inductance accuracy of LDC1101.Please give me some advice
Hello, The LDC1101 measures the frequency and the GUI converts this to inductance based on the capacitance value you enter. Given that there will be part-to-part variability on the capacitor as well as the PCB inductor coil (less significant), the output reading will change from EVM-to-EVM which does not reflect the LDC1101's accuracy. Additionally, the external clock source is provided from the onboard MSP430 so there will be some additional inaccuracies unrelated to the LDC1101. If you want to understand the accuracy of the LDC1101 as well as the resolution, you can refer to the following application note: Optimizing L Measurement Resolution for the LDC161x and LDC1101 [SNOA944]: www.ti.com/.../snoa944 Regards, Luke LaPointe
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with respect to these materials. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.