TDC1000-TDC7200EVM: Delopment guidance
Part Number: TDC1000-TDC7200EVM
So I am trying to test the TDC1000-TDC7200 EVM. I have the 1 MHz sensor that comes with it.
EXP set up.
I have a plastic and a glass container. I am just attaching the transducer using a cellotape for now. I see the following issues:
- Tank level stays at either 2.74 cm or when I reset the TDC1000 at 2.18cm
- The TOF flight reading generally do not correlate with water level. e.g. when I have no water tank on the sensor, it still gives a reading of about 37224 ns.
- When I put a water tank with ~4 cm water, the reading is 37233 ns
Can you suggest what I am doing wrong?
thanks and regards
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Akeem Whitehead:
Sensor Signal Conditioning Applications
In reply to Bharat Aravamudhan:
In reply to Sanjay Gupta96:
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.