Part Number: TDC1011
I am woriking with TDC1011 in a project where 1MHz ultrasonic transducers will be used to measure some media concentration.
However, at this point of the project I replase the ultrasonic signal by a 1MHz sine wave from a GFG-3015 generator. The point is that we are using the TDC1011 only for signal amplification. I mean that we don´t use it to measure the time of flight. We are only interested in the LNA and PGA amplifiers.
So looking to the datasheet, I write a 0x10 value in TOF_1 register focusing by the moment on the LNA (I don´t touch any other register). I see on the oscilloscope a 1MHz sine wave on RX pin, but however no signal is on LNOUT pin. What is going on?
Could someone tell me what am I missing?
Any help will be appreciated.
Thanks in advance.
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Akeem Whitehead:
Sensor Signal Conditioning Applications
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.