This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
Part Number: OPA857
The Transimpedance gain error is 1% typical and 15% maximum. I understand that the 15% take into account all the variability within a production wafer and very wafers.
What kind of typical variation can we expect to have on this parameter for a wafer ?
If you mean the variation in the parameter for the OPA857-DIE, it would be very similar. There are some certain stresses caused by packaging; however, it would not change this parameter drastically.
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to SimaJalaleddine:
Thank You for the answer but it is not what I am asking.
The Data Sheet specifies a Transimpedance Gain variation of typical 1%
with a maximum 15% which is very high compare to the 1% typical.
On any given wafer, what is the range of value of this specific parameter
that we can expect ?
A 15% variation for this parameter seems to be very high taking into account
The fact that it should be controlled by a resistor.
In reply to Tremblay Michel:
The max/typical specs are based on the Gaussian distribution. Within +/- one standard deviation, about 68 percent of the device population will have a value at typical or lower. There will never be a device greater than the max specification. The 15% variation for the parameter is explained in the OPA857 datasheet: "Note that the 2-kΩ internal resistance used for the V-to-I conversion is not trimmed and can vary ±15% with process. Therefore, the source must be capable of sourcing both dc and ac voltages to make sure that the output voltage swing is compliant with the class-A output stage of the TIA block. Any change in the test circuit configuration (such as gain change) requires a new calibration of the internal V-to-I converter."
Thank you for your answer. I understand that the gain variation of +/- 15% applies only when used in the ' Test Mode ' without a photodiode.
If the OPA857 is used with a photodiode in normal use, what is the gain range that we can expect on the Transimpedance gain of 18.2 KOhms and 4.5 Kohms
assuming the load is as specified ?
I believe the same concept would be applied; which would mean the internal resistors that set the transimpedance gain and Rout are not trimmed. This is the reason you would see the same 15% variance, but instead in the transimpedance gain.
Do you have other concerns or questions?
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.