This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
Part Number: INA260
Our customer is looking into INA260 and is wondering what is the lowest current he can measure with it. He would like to use it to measure current ranging from 12 to 35 mA on a 5V Vcc.
Thanks in advance!
This amount of current is not achievable with this device. The reason is that the internal shunt resistor is a fixed 2mΩ, and is made for sensing more broad current ranges. This range will not generate enough current over the inputs to be able to be differentiated well from the input offset which is a worst case 5mA. This offset comprises nearly a third of your measurement, which means that error on the measurement would be >30% error at the lower end. Also, due to the way the device is configured, the smallest LSB size is 1.25mA, so this would be the smallest value that you could differentiate by in the range.
A much better choice here would be the new INA229, since it has lower bias currents for small measurements such as this. It is an external shunt device though, so you will need to choose an appropriate shunt. You could also look at our analog devices for options. I would start with the INA190.
Carolus Andrews, Analog Applications, Current and Hall Effect Sensors
Getting Started with Current Sensing Video Training Series
TI makes no warranties and assumes no liability for applications assistance or customer product design. You are fully responsible for all design decisions and engineering with regard to your products, including decisions relating to application of TI products. By providing technical information, TI does not intend to offer or provide engineering services or advice concerning your designs.
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Carolus Andrews:
Thanks for the prompt response Carolus!
When you said INA229, are you referring to INA229-Q1 or INA2290?
I think INA229-Q1 is not yet fully released.
Thanks to confirm!
In reply to Art Mecina:
I did mean the INA229 here. The automotive version and industrial version of this device are due out at the end of the year. There are preproduction samples of the automotive version (Q1) of the device available now.
If you need something that is fully released as of today, INA226 might work here as well, as its step size is much more appropriate to the range you are looking to measure.
Thanks for the clarification!
I did see the automotive variant of INA229 on TI website but not the Industrial one yet. I will notify the customer on this.
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.