Part Number: LMX2594EVM
We recently purchased 2 units of the LMX2594EVM. I went through the user's guide and was trying to replicate the results shown in figure 9 (phase noise at 14 GHz) to ensure that the units were operating within specification. I downloaded TICsPro and was able to communicate with the eval board without issue. I loaded the default configuration (but changed the output MUX setting from "channel divider" to VCO in order to view the 14 GHz signal at the output. I am using a 100 MHz external oscillator which meets the specifications listed in appendix D at offsets >= 1kHz. I am noticing that the phase noise in the "flat region" i.e offset of 100 kHz that I am measuring is about 2-3 dB higher than what is shown in figure 9 ( -105 to - 106 vs -108 dBc/Hz in datasheet) and about 3 dB higher( -113 vs -116 dBc/Hz in datsheet) at 10 MHz offset. Is this within the normal variability between units? I am measuring the phase noise with a Spectrum Analyzer. I am also seeing a lot of spurious signals especially between 10 kHz and 20 kHz. I noted that cupping my hands over the eval board minimized the level of these spurious signals. Is it expected that the eval board would be so sensitive?In figure 9 of the user's guide the output power is listed as ~ 0 dBm. In my setup my output power at the RFOutA connector is about - 1 dBm.Is what I am seeing to be expected for these units or can you make any suggestions? Thanks for your help.
I think 1dB variation is reasonable. Can you try increasing the reference clock amplitude? The goal is to increase the slew rate which may be a reason of having lower phase noise.
The spurs between 10k and 20kHz, I don't know, could you send me over a screen shot?
Again, 1 or 2dB variation in output power is reasonable, in fact, this could be simply a test equipment issue or because the device gets heat up after running for a long time.
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Noel Fung:
Thanks for the reply. I have limited access right now but as soon as I can I will get a screenshot. Could you recommend a reasonable power level for the reference? I believe I was using a sinusoidal signal at 0 dBm. I didn't see a recommended value in the User's guide for the LMX2594 EVM. From the datasheet for the PLL chip itself, the max ref voltage is listed as 2Vpp. I assumed this was differential into 100 ohms. I am connecting single-ended to the board so I assumed the max voltage would be 0.5 Vp or 4 dBm into 50 ohms. Is this correct?
In reply to PLL101:
0dBm --> 0.63Vpp single-ended. Datasheet is asking for 0.4Vpp to 2Vpp. If possible, you can make it 10dBm.
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.