This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
Part Number: ADS1298
Hello. I'm working in a project with the ADS1298. My question is related to the RC filtering in each electrode input.
In the datasheet of the ADS1298 (page 99, Layout Example) you show differential and common mode filters (screenshot 1), however in the ADS1298ECG-FE User's Guide (SBAU171D, page 52) there are simple RC filters, two filters in cascade.
In the past, I implemented the datasheet scheme, with the differential and common mode filters with an aprox. 150hz cut off frequency to limit the signal at the bandwidth of interest for an ECG.
Now I'm working in a new implementation and saw the filtering used in the evaluation module, and came across with the question of why that filters have a higher cut off frequency (153 kHz with the 22.1k and 47pf, and 338 kHz with the 10k and 47pf) and, why you used that filters and not the ones in the datasheet?.
Thank you for your post!
The RC filters in the signal path are primarily used for antialiasing purposes. It's not necessary to set the cutoff frequency exactly at the end of your bandwidth of interest. The goal should be to attenuate signals near the modulator sampling frequency for delta-sigma ADCs. I've written a blog series on the subject to help explain this concept, part 1 of 3 is linked below:
This EVM released some time ago and through experience with other customers, we've learned that some use both common-mode and differential RC filters in their designs. A differential filter will help maintain good CMRR, which is a key spec in ECG applications. We recommend setting the differential filter cutoff frequency at least one decade less than the common-mode filter cutoff frequency.
Applications Engineer | Precision ADCs
Are you working on a bio-potential application? Check out these helpful resources: ADS129x BIOFAQ | ECG Online Training
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Ryan Andrews:
Thank you Ryan!, that answers my question really clear.
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.