This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
Part Number: ADS1294
We are using the ADS1294 for ECG data acquisition we are facing the below-listed Problem.
1. We are feeding 50Hz Signal to ADC and when we are plot FFT of the signal in MatLab it is showing 51Hz same thing happens with other frequency. All frequency whatever we are feeding is shifting by 1Hz from given Frequency.
ADC Sampling Rate: 500SPS
Thanks for bringing this to our attention!
Is there any variation to this or is it always +1Hz?
Are you using the PDK or is this part of your own design?
It is possible that your signal generator is the source of the error, have you verified the performance of the signal generator using a more precise piece of equipment?
Alex SmithApplications Engineer | Precision Delta-Sigma Converters
Check out our helpful resources: BIO-FAQ: Common Questions for TI's ADS129x Family of Bio-Potential ADCs
TI Precision Data Converters | TI Precision Labs - ADCs | Analog Engineer's Calculator | Data Converters Learning Center | Selection Guide
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Alexander Smith:
==>It happens always with all signal
==>Part of Own design
==>We have checked all the parameters of the source from where we are giving the input signal to ADC.
After long troubleshooting, we found out the internal oscillator is generating the issue we enable the external 2Mhz oscillator and this thing works fine now there is no frequency shift in ADC output.
Very disappointed with Texas Instruments. We almost waste 3 months to resolve this issue. Please check and verify your ADS1294.
In reply to MUKUND PIPALAVA1:
I greatly appreciate you bringing this to our attention. I will make sure this feedback is known by the design team.
I would like to get a little more information from you in regards to this issue.
Is the device ID returning correctly, and can you confirm that you are sending a /RESET pulse or command after power-up?
Does this issue apply to all data rates, all OSR combinations, and high performance and low power modes? Does the 1Hz difference scale as you change the data rate and OSR combinations, or is it always an absolute 1Hz?
How many points are being collected for the FFT? It is possible that the bin size is too small to resolve the correct input frequency.
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.