Part Number: ADS1299EEGFE-PDK
Hi TI's expert,
I downloaded the ADS1299EEGFE-PDK-SW and used the ADS1299EEGFE-PDK boad to test the RLD function, everything works well except that when I wanna enable the Bias sense function ,I can't find a choice on the software to configure it. I tried to change all the choices on the software but the reg3 bit 1 just keep read as 0, Also the Debug-Manual register write seems didn't work（do not know how to use it,click on the right button with 03,seems no response,or else I can mannually write 1to that bit ） See below pics.so I can't enable the RLD function ,the ECG wave shows a large 50HZ common mode noise.
BTW, In our own ADS1299 board we can write the register 03 bit 1 to 1 and it seems working well.
What can I do to enable that ?or do you have a new version of the sofware?
Thanks very much
Thank you for your post.
The BIAS lead-off sense function is not available on the EVM. By default, this bit is set to 0. This function is only meant to check the connection status of the BIAS electrode. In the actual application, you will have to temporarily power-down the BIAS amplifier to check the status, then enable the amplifier again to resume normal measurements.
The BIAS lead-off sense function is independent of the main function of the BIAS electrode, which is to establish the common-mode voltage on the body and cancel the common-mode noise measured by the input electrodes. You can continue to use the BIAS electrode to reduce common-mode noise without this lead-off function. Please refer to our E2E BIOFAQ page for some useful collateral on using the BIAS electrode (also referred to as RLD electrode) to improve CMRR.
Applications Engineer | Precision ADCs
Are you working on a bio-potential application? Check out these helpful resources: ADS129x BIOFAQ | ECG Online Training
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Ryan Andrews:
Thanks for your prompt reply.
1.From sba188 we know that the fBW caculation:
and in the ADS1299_EVM the RF CF are(RF=392k,CF=10NF),then fBW=1/(2*3.14*392k*10n)=40.6Hz,is it a Low Pass Fitler?Then how can it cancel the 50Hz CM noise?
In the datasheet the value are 1Meg and 1.5nF then the fBW will be more than 100Hz,I think that is more rational. what value do you think we should use in China(220v/50Hz)?
2. The datasheet recommand we using the differential cap between P and n in the inputs,but the EVM uses the shunt cap to GND,I used both methords in our own board(5V singal supply differ from its bipolar+2.5/-2.5 Power suplly ) and found that the shunt methord's CM noise it less than the differential one.what should I choose?
In the datasheet:
in the EVM:
In reply to tinky1999:
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.