Part Number: DAC712
1. DAC712U has been used for more than 10 years in my customer device,recently, my customer purchase 240pcs DAC712U, test 12pcs, 1pcs DAC712 Fail
iI is normal while DAC712u work ten minutes before, but afer ten minutes , output voltage deviation more than %0.4
2. Shutdown power ,restart the device , but it is keeping output voltage deviation more than %0.4
3. cross-over experiment, details data show as below
DAC712u test data
1#ok is qualified board and 2#NG is defective board
from test data，we can see that the output voltage deviation phenomenon follow DAC712U any advice？thanks in advance~
0.4% error is not acceptable for DAC712U. Total unadjusted error for the device is approx. +/- 20mV ( includes INL, DNL and Gain error).
I am curious to note the measurements for the Vref . Min and Max for reference voltage is 9.975 to 10.025V, where as in your case you are measuring higher voltages. How are you measuring these voltages? Is this measurement done with precision multimeters?
Please note that the LSB size of the device 305uV assuming a 10V reference and +/-10V output range.
So for a code of -32088 should produce ideally -9.786V +/- 20mV ( Assuming offset error is nullified).
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Akhilesh K:
Direct test voltage on pin5, using Agilent 34401A precision meter measurement.
What is the reason for the reference voltage of 10.17v and 10.22v?
After testing the A-B-A Switching IC, we find that the problem follows the IC. Are the parameters of the new IC different from those of the old one?
In reply to HY yang:
This reference voltage looks interesting.
Need to check with production team for any change in test program revision or limits change, Vref of 10.17 is clearly outside the limits of the specification.
I am assuming you are using a calibrated meter. Also can you please check the reference voltage by disconnecting the trim pots from the Vref?
I just want to isolate the reference voltage alone.
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.