This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
Part Number: DLP2010NIR
I had Halogen Lightsource HL-2000 by my side, with a Thorlabs filter at 1550nm and FWHM=12nm. I tried registering spectrum with NIRscan Nano, and I can not get an FWHM at 12nm. I used the parameters used in the document dlpa066.pdf :Start wavelength— 900 nmEnd wavelength— 1700 nmPixel width of patterns— 5 pxNumber of wavelength points— 427Number of scans to average— 50Scan type— column
I measure FWHM=16nm. I managed to go down to 15nm but no more, with :
Start wavelength— 900 nmEnd wavelength— 1700 nmPixel width of patterns— 2 pxNumber of wavelength points— 623Number of scans to average— 50Scan type— column
What would be the parameters to modify to obtain an optimal resolution of 10, as indicated in the user manual?
Thank you, and have a nice day !
Please allow us to look into this and we'll get back to you. In the meantime could you please confirm the firmware version number you are using?
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Kyle Rakos:
I use the firmware version v1.1.9 (NIRscanNano GUI)
In reply to lea bonhomme:
Hi Lea, Could you kindly update the NIRScan Nano GUI to v2.1.0, NirScan Nano TIVA SW to v2.1.0 and DLPC150 FW to v2.2.0? You are always advised to use the latest version of SW and FW to obtain the maximum reliability and accuracy from the device. Also, could you kindly try taking a reference scan using an uniform spectralon surface before taking sample scans? This also improves the accuracy of the obtained data.
Thanks & Regards,
Thanks & Regards,Hirak.
If a post answers your question, please click on "Verify Answer" button
In reply to Hirak Biswas:
I update the NIRScan Nano Gui to v2.1.0,NirScan Nano TIVA SW to v2.1.0 and DLPC150 FW to v2.2.0. I take a reference scan with a spectralon surface, and i obtain a FWHM = 13,4 nm.
According to the Thorlabs data sheet, the filter has a FWHM = 12nm +/- 2,4nm, so I am within tolerances. I do not know if the difference comes from the quality of my filter or my settings, but it is already better.
Thanks and Regards,
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.