This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
Part Number: MSP430FR6043
Good day. Our company develops new gas meters and we want to use your controller MSP430FR6043. We have several questions: 1. When the values of the time difference between the flow and against the flow (Delta TOF) of about 20,000 ns, there is a limitation in the calculation of flow. Is it possible to expand the range? 2. Is it possible to set the frequency of the probe pulses (Pulse output frequency) to 100 kHz? We can’t to set this frequency using the USSWLib library.
Please find answers below:
1. The maximum dToF shift can be increased by increasing USS_ALG_MAX_SAMPLE_SHIFT in the USS_userConfig.h file from 20 to a larger number.
2. The minimum supportable frequency of the FR6043 PPG is 133kHz.
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Leonardo Estevez:
Hello Leonardo!Thank's a lot. Your answer to the first question was useful. The minimum possible value of the PPG frequency that we were able to set is 160kHz. Could you tell us what we are doing wrong?
In reply to Andrey Dolya:
You need to reduce the HSPLL frequency to 68MHz. This will also give you a lower sampling frequency.
Hello Leo!The USS_HSPLL_FREQ_IN_MHZ parameter in the USS_userConfig.h file was set to 68, but this was not enough. What have I done wrong? What more needs to be done?
If you are running from the header files, you can't update via the GUI and expect the same parameters to be there. You should export the header files from the GUI into your project and then change the HSPLL setting before rebuilding. You can then connect to the GUI and capture an ADC signal to confirm you are getting the lower sampling frequency. Please note that the sampling frequency should also be reduced from 1000 to 850.
I noticed that you are using a previous release of the GUI. Please download the latest version of the software and GUI(and reflash your board) - these can be found here:
If the GUI version is out of sync with the image on the EVM, the generated header files may not work properly.
I did everything as you said. I installed a new version of the software. Now it is not possible to set the parameter (signal sampling frequency) to 850.
Any more suggestion for this question? thanks.
In reply to KC Xu:
We don't currently support this lower sampling rate in our GUI. I'm working with our developers to determine when this might be changed.
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.