Remote Temperature Sensors — What is n-factor?
This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
The η-factor or ideality factor, of a Bipolar Junction Transistor (BJT) can be considered to be the measure of how closely a BJT follows the ideal equation (shown below). An ideal BJT has a η-factor = 1.000, but in a non-ideal world this is not the case. Most remote temperature sensors default η-factor setting is 1.008 or 1.000 depending on the remote temperature sensor. If the remote BJT’s η-factor does not equal the default value then the n-factor correction register will have to be used to compensate for this discrepancy.