This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
I have constructed an LVDT that is not behaving as it theoretically should. I have eliminated so many possibilities for its malfunction that I thought I would post on here to see if anyone would help. I have been running COMSOL simulations with altered coil numbers and geometrical asymmetries to see if I could reproduce the behavior with no success.
I've wound a 3D printed form with excitation (primary) coils (30 turns) and pick up (secondary) coils (2 unconnected sets of 30 turns which add to 60 turns). It's a large LVDT, the radius is .10008 m and the length is .09906 m. I have 1.035 A running through it
Here is what is happening: I have a 1 inch steel ball that I use as the core, as I move it from the right hand secondary coil to the left side coil, there is a huge response at the right hand secondary coil, then once the ball passes the null position the response doesn't change much except show a phase shift, the signal actually decreases a bit at the left coil. The amplitude response seems to be concentrated on the right coil and the phase response concentrated on the left.
Can anyone explain this strange behavior? I'm a graduate student at NASA Langley and my advisor can't explain what's happening, nor can the electronics expert.
Any help or assistance would be very appreciated.
Would you please provide a schematic of the electrical components of this design? Please include all power sources, electrical connections to the coils, etc. Reading your description I'm not understanding what electronic circuitry is involved and specifically what ICs are in use. Once we see the circuitry involved, we will better be able to assist.
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to LeonardEllis:
Hi Leonard, above is the schematic I drew including all the ICs, power supplies, inputs, and passive components. Please let me know if I can elaborate on anything else.
Thank you so much for taking the time to help.
In reply to Michelle Weinmann:
Hi Michelle, I'm not an expert on LVDTs, but I'll try to help you out and see if the TL082 is working as desired. Can you provide some oscilloscope shots (or other empirical data) of the following: * Input signals for the TL082 when the ball is left, right, neutral (to compare against each other) * Output signals for the TL082 when the ball is left, right, neutral It would also be useful to see if the signal that is driving the primary coil stays the same as the ball is moved into the various positions. Thanks, Paul
In reply to Paul Goedeke:
Thank you so much for offering your help. I've attached all the signals from the scope that you requested. The signal driving the primary coil does stay the same when I move the ball around, so at least that works!
Please let me know what you think.
in, left in, neutral in, right
out, left out, neutral out, right
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.