This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
I am building a digital wattmeter intended for measurements on AC power lines. Many designs from Ti Designs and various Application Notes employ a similar frontend where the mains voltage is first scaled down using a voltage divider. The scaled-down voltage signal is then taken as differential and passed through some low pass filter before being fed to a differential-input amplifier or directly to a high-resolution differential-input ADC.
( simplified diagram of the input circuit from SimpleLink™ Wi-Fi® CC3200 Smart Plug Design Guide www.ti.com/.../tidu983.pdf)
After running a simulation of this input circuit, I have found that the ground connection on the Neutral line renders V_SENSE- signal thousands of times smaller than the V_SENSE+ signal, and causes a non-180° phase difference between them. On the contrary, if the connection is removed, V_SENSE+ and V_SENSE- signals are of the same magnitude and 180° relative to each other, creating a perfect differential signal.
1. What is the purpose of this connection from Neutral to the reference ground?
My second confusion is also related to the above-mentioned input section, in a slightly varied form, which appears in many Ti Designs and application notes.
( simplified diagram of the input circuit from Smart Plug with Remote Disconnect and Wi-Fi Connectivity www.ti.com/.../tidu531.pdf)
This is largely the same as the previous configuration. Apart from the differences in components values, the most intriguing difference is in R5, which is 100Ω as opposed to 1kΩ for R6. According to the document, the justification for the smaller value is to make up for the much larger impedance at V_SENSE+. However, simulation results show that the outputs are almost identical to those of the previous configuration. Moreover, the magnitude mismatch and phase angle problems still exist.
The same configuration also appears inwww.ti.com/.../tidu454a.pdfwww.ti.com/.../tidu453.pdfwww.ti.com/.../tidu816.pdf
2. What is the purpose of using a smaller value for R5? Why is it that the previous configuration use the same value for R5 and R6? How do I know when to use which?
Thank you very much :)
Thank you for your questions; I have alerted the designer of this circuit to reply. Apologies for the delay.
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
To better aid the community, please click on the "This Resolved my issue" button whenever a post answers your question!
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.