Other Parts Discussed in Thread: DRV5033
My customer is verifying the DRV5033 with the N52 magnet specifications below. Should we enter the parameters of Magnet Materials (N52) of Remanence (4000 Gauss) in the design tool at http://www.ti.com/product/DRV5033/toolssoftware? WHat is product line's recommended equipment setup to verify the flux?
We got different distance result as the pictures below (6.3mm vs 3.58mm). Customer confirmed that the 3.58mm does not work with actual experiment.