This thread has been locked.
If you have a related question, please click the "Ask a related question" button in the top right corner. The newly created question will be automatically linked to this question.
Part Number: DRV5032
Do you know what equipment can measure DRV5032 BOP?
Can you please be a little more specific on the request ?
A known magnetic field is required to measure the parameters. As long as a magnet with a known magnetic field is placed in a controlled environment the device can detect the magnetic field and it would trip.
What are you sensing today ?
Hall Effect Sensing FAQ Inductive Sensing FAQ Ultrasonic Sensing FAQ
"TI makes no warranties and assumes no liability for applications assistance or customer product design. You are fully responsible for all design decisions and engineering with regard to your products, including decisions relating to application of TI products. By providing technical information, TI does not intend to offer or provide engineering services or advice concerning your designs."
We are glad that we were able to resolve this issue, and will now proceed to close this thread.
If you have further questions related to this thread, you may click "Ask a related question" below. The newly created question will be automatically linked to this question.
In reply to Arjun_Prakash:
Customer would like to know how to measure BOP?
What test equipment can measure BOP?
In reply to Ian Wang:
In order to measure BOP, it is necessary to operate with a fixed magnetic field at a specific location. It is best done by using a Helmholtz Coil driven by an accurate source in conjunction with a precision system for setting the absolute position of the sensor relative to the coil. When doing this it is critical to use the published location of the sensor internal to the package for best results. The source can be driven to sweep the magnetic field until BOP is triggered.
A less precise method is to use a magnetometer to detect the field component the sensor will detect from a permanent magnet and to place the sensor in the same location. This requires being able to place the sensor of the magnetometer in the same location as the Hall Sensor and is prone to setup errors.
Another alternative is to find the BOP trip point by moving the magnet near the sensor, and then use an online calculator tool to determine what the field components are at the detected location.
All content and materials on this site are provided "as is". TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. No license, either express or implied, by estoppel or otherwise, is granted by TI. Use of the information on this site may require a license from a third party, or a license from TI.
TI is a global semiconductor design and manufacturing company. Innovate with 100,000+ analog ICs andembedded processors, along with software, tools and the industry’s largest sales/support staff.